Pages containing 'results'
Is there anyway to make things line up nicer in the results after a MET/CAL procedure finishes?
For example, I have a performance verification procedure I wrote for load cells and here is some sample output:
I am fairly new at writting Met/Cal and I am having a little trouble. I wrote a procedure to take a voltage value and turn it into Watts and calculate the % of error. Everything runs great but the resolution of the results is too large for all report values. Below is the sample of the code. Can anyone tell me if I can set the resolution of the Math FSC to a certain value such as .0001 or whatever I need?
MET/CAL requires certain information before it can calculate and report measurement uncertainties. When using an Instrument FSC, the minimum information required can be found in this sample procedure for a handheld digital multimeter:
Ensure that your procedure code does not make use of scientific notation. For example, this section of code below instructs MET/CAL to generate scientific by defining the range as “5e4” in step 1.007: